Class: Instrumentation Manufacturer: Kajaani Model No: SD Part Number: Qty: 1. Description: Automatic Sample Valve. follow-up through a communicator or Kajaani interface via Ethernet. Three operating modes: • Manually initiated. • Collection timer 24/7. STEC SEC Mass Flow Controller. ₪. METSO KAJAANI SD TRANSMIITER USED. ₪. MKS Flow Meter (N2) – Model ALS3BV.
|Published (Last):||22 June 2010|
|PDF File Size:||6.98 Mb|
|ePub File Size:||1.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Add to watch list ,ajaani from watch list. Non-recyclable materials Non-recyclable materials must be disposed of according to currently valid local legislation and regulations. Metso will accept only thoroughly cleaned devices, and the cost of recycling will be invoiced from the customer; please ask for recycling quotation.
KURZ 454FTB-16-HHT Single Point Insertion Thermal Mass Flow Transmitter
Maintenance every 12 months? Have one to sell? Hot pulp ejected with high pressure may cause serious accidents. Cs Process pressure 0—4 bar 0—58 psi 0. For pulps containing chlorine compounds, when the process consistency is 0. This page was last updated: Tighten the hex socket screw properly. Turn the nut part 21 back towards the other nut, leave it rather loose at this point. Screw the new pressure kahaani part 6 to the bushing part Turn the tool by its arms and screw it into the bearing bush as shown in Fig.
Construction of sample delivery valve.
kajaani in Business & Industrial | eBay
Make kajaanl to use appropriate safety equipment and to clean the device as necessary before recycling. You may need to open and close the valve several times before it begins to operate smoothly. SD installation, operating, service – 4. Best Match Best Match.
Adjust with a tool, NOT with your hands! Close the air supply to the sampling device by the manual valve. Skip to main content. Wind the piston shaft part 15 loose from the actuator using two wrenches spanner gaps: Unscrew the pressure switch part 6 from the bushing part Save kajaani to get e-mail alerts and updates on your eBay Feed. Loosen the locking screw hex socket screw M6, part 7.
ROSEMOUNT P D1A1E5B4M5DA1 4-Wire Display Temp. Transmitter NEW IN BOX – $ | PicClick
Connect instrument air pressure about 6 oajaani, 87 psi to the actuator, coupling d. This ensures that no objects remain between the sealing surfaces and prevent the piston from closing completely.
Removing the seal ring: Do not put your fingers near the blade! After adjustment, close the cover carefully.
Metso Field Systems Manufacturer reserves the right to changes without prior notice. Using the push bar Fig. Replacing the bearing bush and O-ring This operation requires a process shut-down — the sampling device must be removed from the process. Kajani will be charged to your account from the purchase date if the balance is not paid in full within 6 months. Positioning the flow inlet opening In Fig.
NEW – Siemens DMT-2000 Circor Liquid and Gas Multivariable Flow Meter
Remove the 4 hex cap screws that hold the ball valve together, and oajaani actuator will come loose; see Fig. Disconnect the water inlet hose and sample line from the sampling device Fig. The flow opening of the piston opens more when the piston moves a longer distance.