Does anyone have any information on how to calibrate a Hickok model tube tester? I’ve searched, but can’t seem to find any links. Thanks!. HAPPY HOLIDAYS TO ALL, Got a question about a Hickok I’ve got a chance to purchase one of these locally but in the picture I’m sending. After waiting years too long, I finally bought a really nice looking, lightly used Hickok and it was doing funny things with known good tubes. It read shorts until.
|Genre:||Health and Food|
|Published (Last):||24 November 2010|
|PDF File Size:||9.25 Mb|
|ePub File Size:||19.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
A recommended list of Hickok tube testers
I do have a Hickok a I guess I could switch meters just for testing purposes, that is if I buy it. Avoid like monkey plague. The problem I have is that it doesn’t make sense where that loose wire came from or goes to.
These documentations are only intended for qualified hickook who are aware of the respective safety regulations. One other thing is this As Dave says, it sounds like somebody was playing “Nick, knack, paddywhack” inside there with the innards.
Dynamic Mutual Conductance Tube Tester 536
These terms and conditions are subject to change at any time and without notice. But they ARE the achille’s heel of old Hickoks. I can have your old meter rebuilt hicklk proper specifications. Other offers may also be available. Models and A. Apr Sun 27, 3: Hi all, After waiting years too long, I finally bought a really nice looking, lightly used Hickok and it was doing funny things with known good tubes.
Any help would be deeply appreciated!
See other items More Interest will hiclok charged to your account from the purchase date if the balance is not paid in full within 6 months. The B and C units have a “VR” function to give a quantitative test on voltage regulator tubes like 0A2. Data is widely available. All meters are new. It could simply be misadjustment of the zero setting on the meter, but that seems unlikely.
Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Thanks for your help!
Have one to sell? Tap gently on the meter to see if it returns. I finally got a schematic to download from BAMA on my son’s PC, and it’s a bear to determine the wiring of those resistors from the nomenclature of the document I have.
Which begs a question from me: Data is widely available for the B, C units. Was one end of a resistor disconnected when you got hcikok Contact the seller – opens in a new window or tab and request a shipping method to your location.
Looking at the schematic and trying to identify which resistor is ‘R28’ or whatever led me to try drawing a layout with mixed success Notify me of new comments via email. The A does not have 4,5,7 pin 5536 sockets, but it does have the 9 and 12 pin compactrons.
Operating Manual For Hickok Tube Tester Original Plus Other Papers | eBay
Start Manufacturers Categories Search Upload a manual. Any international shipping and import charges are paid in part to Pitney Bowes Inc. I’ve never owned a Hickok tester until about a month ago which I picked up a a. Shipping is additional and calculated by order.
AC line and DC bias meters also available. Seller does not offer returns. I’m pretty sure I’ve answered my own question, but the manual and the Hickok guy seem to indicate otherwise.
People who viewed this item also viewed. My new old Hickok has an open spool resistor and. These hicko nice compact testers and data is widely available. The TV-3 is a very cool tester.